4Long time monitoring BP and SpO2, displaying value of BP, SpO2 and PR. 5NIBP and SpO2 data Record for large capacity. 6With friendly user interface, the user can see list menu and review measurement results. 7The device can display low power information, alarm information, error information and time information richly. 8Parameter alarms function is optional.
9Patient information can be flexibly set. PC software can achieve data review, analysis measurement results, seeing trend, printing reports and other functions. 11Data can be transmitted between device and computer, such as upload patient information, download measurement data. Function of master device software: 1Set patient information through PC software.
2Up to 1000 patients cases can be edited and supervised. 3The time segment of dealing with the patient's NIBP data is 48 hours. 4Connect the device by USB interface. 5Can upload patient information data collection project and download collection data.
6Can display scoop-shape trend graph, filling-type trend graph, histogram, pie chart, correlation line graph. 7Can edit every piece of data, and add annotation to it. 8Can edit basic information, doctor's advice information, NIBP and SpO2 status instruction, current medicine-taken information, etc. 9Support print preview, print the report.
NIBP: Measurement Method: Oscillometry Measurement Mode: The upper arm measure Measuring range: SYS 40mmHg~270mmHg MAP 20mmHg~235mmHg DIA 10mmHg~215mmHg Automatic Measurement Interval: 126.96.36.199.188.8.131.52.120 minutes Resolution: 1mmHg Accuracy: ±3mmHg Increasing pressure mode: force pump increases pressure automatically Reducing pressure mode: self-motion ladder reducing pressure mode Alarm parameter: SYS / DIA SpO2: Measuring range: 0%~100% Resolution:1% Accuracy: 70-100%, ±2%, below 70% unspecified Error in Weak Filling Condition:SpO2 and pulse rate can be shown correctly when pulse-filling ratio is 0.4%. SpO2 error is ±4%, pulse rate error is ±2 bpm or ±2% (select larger). Resistance to surrounding light:The deviation between the value measured in the condition of man-made light or indoor natural light and that of darkroom is less than ±1%. Pulse Rate: Measuring range:30bpm250bpm Resolution:1bpm Accuracy: ±2 bpm or ±2% (select larger) Display: 2.4" TFT colour LCD Safety: Power: DC 3V(Two "AA, 1.5V Alkali Battery) Product safety type:Type BF application part (Internally powered, defibrillator protected). Cuff for adult Integrative SpO2 Probe USB data line Disk (PC software) Pouch User manual.Dimension: 1286936 mm (No including Packing) Weight. ComIndex code:check on the package, any problem please contact me. Food and Drug Administration and state and local regulatory agencies. The Fingertip Pulse Oximeter is registered on the Australian Register of Therapeutic Goods (ARTG) with the code 197923, and certified by FDA of United States and CE, TUV of Europe. The Fingertip Pulse Oximeter that is FDA 510K Approved. 100% satisfaction is our goal!
We are looking forward to establishing a successful business relationship with you. Only english user guide, if you need any other language, please contact me. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Medical & Lab Equipment, Devices\Blood Pressure Machines & Monitors". The seller is "contecstore" and is located in this country: CN.
This item can be shipped to North, South, or Latin America, all countries in Europe, all countries in continental Asia, Australia.